[time-nuts] 5061 Cs frequency shift with long time constant?
phk at phk.freebsd.dk
Tue Apr 5 12:49:34 EDT 2005
In message <4252B46E.3010509 at pacific.net>, Brooke Clarke writes:
>> I got my 5061A trimmed to the point where a 5 day run against GPS
>> showed an offset of less than 1x10e-13. I then switched the unit into
>> the long time constant mode and now see that the offset (over 36
>> hours) is -1.7x10e-12. Nothing else in the setup has changed.
Yes, very likely. The leak-current in the integrating capacitor cannot
be fully ignored at this level.
Poul-Henning Kamp | UNIX since Zilog Zeus 3.20
phk at FreeBSD.ORG | TCP/IP since RFC 956
FreeBSD committer | BSD since 4.3-tahoe
Never attribute to malice what can adequately be explained by incompetence.
More information about the time-nuts