[time-nuts] Fw: HP 105 powersupply problems
robert.atkinson at genetix.com
Tue Apr 4 03:23:23 EDT 2006
Using a Variac over a long period of time is great with conventional
(Transformer-rectifier-capacitor) linear PSU's, but be careful with more
modern equipment with switchmode PSU's. They can draw excessive current
during undervoltage conditions.
Still use your Variac, but ramp it up a little quicker, maybe 30s to a
minute. Watch out for auto supply voltage (115/230) switch designs too,
it's best not to cross the switching boundary too slowly as they may not
switch back from voltage doubling to full-wave rectification resulting
in very excessive internal voltages.
If you don't have a Variac, try a low wattage (40W) light bulb in series
with the supply to limit the current.
From: time-nuts-bounces at febo.com [mailto:time-nuts-bounces at febo.com] On
Behalf Of Poul-Henning Kamp
Sent: 04 April 2006 06:28
To: Discussion of precise time and frequency measurement
Subject: Re: [time-nuts] Fw: HP 105 powersupply problems
In message <002901c6578c$9d928120$0e8fae43 at hsd1.in.comcast.net>,
"Richard and M
arcia Putz" writes:
>I had similar problems with my HP105, not bad transistors, check
>out the filter capacitors. Electrolytics in any equipment past 15-20
>years of age could be suspect. I'd look there first.
In general when powering up equipment that has been turned of for
years: Use a variac and ram the voltage up slowly over a couple of
Electrolytics can loose their polarisation over time and if exposed
to full voltage they are likely to splatter all over the place, but
if you give them a slow start they will most often recover their
polarisation and work fine.
Poul-Henning Kamp | UNIX since Zilog Zeus 3.20
phk at FreeBSD.ORG | TCP/IP since RFC 956
FreeBSD committer | BSD since 4.3-tahoe
Never attribute to malice what can adequately be explained by
time-nuts mailing list
time-nuts at febo.com
Any opinions expressed in this email are those of the individual and not necessarily Genetix Ltd (Genetix) or any company associated with it. This email and any files transmitted with it are confidential and solely for the use of the intended recipient. If you are not the intended recipient or the person responsible for delivering to the intended recipient, be advised that you have received this email in error and that any use is strictly prohibited. If you have received this email in error please notify Genetix by telephone on +44 (0)1425 624600.
The unauthorised use, disclosure, copying or alteration of this message is strictly forbidden. This mail and any attachments have been scanned for viruses prior to leaving the Genetix network. Genetix will not be liable for direct, special, indirect or consequential damages as a result of any virus being passed on, or arising from alteration of the contents of this message by a third party.
More information about the time-nuts