[time-nuts] Telecom Solutions DCD Cesium
Lutwak at Alum.mit.edu
Fri Mar 17 21:17:37 EST 2006
Try downloading Monitor3 from the Symmetricom WWW site:
Connect your PC to the cesium module with a null modem cable, typically
DB-25 on the cesium end.
Start by trying either 2400-7-O-2 or 9600-7-O-2.
If that doesn't start talking, try running "Detect ComPort settings" on the
"File->Configure Serial Port" menu.
Monitor3 should work with most of the Datum/FTS 5045/4201 cesium modules.
The control functions don't work, but it should be able to read all the
monitor parameters, save them, etc.
Robert Lutwak, Senior Scientist
Symmetricom - Technology Realization Center
34 Tozer Rd.
Beverly, MA 01915
(978) 232-1461 Voice RLutwak at Symmetricom.com (Business)
(978) 927-4099 FAX Lutwak at Alum.MIT.edu (Personal)
(339) 927-7896 Mobile
> -----Original Message-----
> From: time-nuts-bounces at febo.com [mailto:time-nuts-bounces at febo.com] On
> Behalf Of Magnus Danielson
> Sent: 17 March 2006 17:20
> To: time-nuts at febo.com
> Subject: [time-nuts] Telecom Solutions DCD Cesium
> This morning a Telecom Solutions DCD Cesium landed on my lab-bench at
> There is something wrong with it, but there should still be some time left
> on the tube according to the source. Anyway, I am looking for info (the
> *should* be on the way, but I will not hold my breath) and especially
> anything giving away usefull info on the serial port as well as service
> Any hints and tips is welcome.
> In the meanwhile I am figuring out how to solve the 48V problem, but I
> a cludge available.
> time-nuts mailing list
> time-nuts at febo.com
More information about the time-nuts