[time-nuts] On some pitfalls of the dual mixer timedifferencemethod of h...
Dr Bruce Griffiths
bruce.griffiths at xtra.co.nz
Sat Oct 7 22:59:36 EDT 2006
Tom Van Baak wrote:
>> Interesting as the techniques employed may be, the resultant performance
>> is still inadequate for characterising state of the art oscillators.
>> The instrument phase noise floor is higher than that specified for
>> current low noise (but not necessarily low drift) crystal oscillators.
> Well, true, but perhaps one definition of "state of the art"
> is simply when the high-end test tools that you can buy
> off the shelf are inadequate compared to the newest
> technology you are building in the lab...
> time-nuts mailing list
> time-nuts at febo.com
My point really was that "inferior" analog techniques such as 2
reference source 2 mixer cross correlation analysers have a phase noise
floor 15-20dB lower than the supposed superior direct digital mixing
techniques. Such cross correlation instruments are commercially
available from Wenzel associates and others.
The latest Agilent phase noise analysers use analog mixers, dual digital
phase locked loops and digital cross correlation.
More information about the time-nuts