[time-nuts] How to measure Allan Deviation?
John Ackermann N8UR
jra at febo.com
Sun Oct 22 21:15:12 EDT 2006
Dr Bruce Griffiths said the following on 10/22/2006 07:33 PM:
> If you are going to use a PPS divider to divide the oscillator frequency
> down to 1Hz, you will need to measure the inherent jitter of the divider
> to ensure that it doesn't degrade the measurement resolution. It may be
> necessary to resynchronise the divided output using a fast D flipflop to
> reduce the inherent divider jitter to less than the 20ps resolution of
> the 5370.
The few experiments I've done indicate that there isn't that much
sample-to-sample jitter in a divider built up out of a reasonable-length
string of 74HC390 dividers, but the temperature stability is horrible.
More information about the time-nuts