[time-nuts] How to measure Allan Deviation?
Dr Bruce Griffiths
bruce.griffiths at xtra.co.nz
Sun Oct 22 23:52:33 EDT 2006
Didier Juges wrote:
> Dr Bruce Griffiths wrote:
>> In principle this measurement could be made with a time interval counter:
>> PPS -> START
>> delayed 10KHz -> STOP
>> Vary the delay and watch the jitter jump when the leading edge of the
>> PPS signal occurs during the 10KHz burst which was phase coherent with
>> the previous PPS pulse.
>> The only problem is finding a suitable variable delay device with
>> sufficiently low (<=1ns??) jitter.
> The delay device can be triggered by the 1 PPS, then will drive the ARM
> input of the counter, so as long as the delay device's jitter is less
> than the 10 kHz period, if we adjust the delay to 0.99985 second
> (between the last 2 periods of 10 kHz before the 1 PPS), then the TI
> counter will START on the last 10 kHz pulse before the pps, and STOP on
> the 1 PPS.
> Tek has some time delay generators in the TM-500 and 7000 plug-in
> series. I knew one day I would need one of those, I now know why :-)
> time-nuts mailing list
> time-nuts at febo.com
Since the delay generator is clocked at 10kHz 4029B synchronous up/down
binary/decade counters should be satisfactory.
Since the edge speeds are relatively slow breadboarding will be somewhat
simpler and more reliable.
More information about the time-nuts