[time-nuts] May I ask for your advice on a faulty 5370B?
Dr Bruce Griffiths
bruce.griffiths at xtra.co.nz
Tue Jul 3 01:52:42 EDT 2007
S McNamara wrote:
> G'Day All,
> I just purchased a 2nd hand HP5370B counter.
> It was last calibrated in 2002 but fails the operators checks.
> I have run tests 1-21 and only a few fail.
> In step 13 the display indicates 90.9*ns
> In step 15 the std dev is 20.* to 16.*ns with a min of 90.9*ns and a max
> of 91.*ns
> In step 17 the stop period is 91.*ns and the start period is 8.91ns
> I have measured its ref output with another counter and it seems stable
> and 10MHz.
> So I am seeking ideas on where to start to fault find, any guidance
> would be much appreciated.
> Regards Scott
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
Check the frequency multiplier chain alignment.
More information about the time-nuts