[time-nuts] Cs stability
rk at timing-consultants.com
Tue Jul 17 08:40:17 EDT 2007
Another way that has been incorporated successfully to combat g sensitivity
are accelerometers mounted on each of the x, y, Z axes of the crystal.
Combining this with spring mounts works well. g sensitivities down to e-10
per g can be achieved.
From: time-nuts-bounces at febo.com [mailto:time-nuts-bounces at febo.com] On
Behalf Of SAIDJACK at aol.com
Sent: 17 July 2007 06:14
To: time-nuts at febo.com
Subject: Re: [time-nuts] Cs stability
); SAEximRunCond expanded to false
time-nuts-bounces+rk=timing-consultants.com+rk=timing-consultants.com at febo.c
In a message dated 7/16/2007 12:10:54 Pacific Daylight Time,
SAIDJACK at aol.com writes:
>For example tilting the Cs unit by 90 degrees will typically give an
>of about 1E-09 or so.
>It would take some time for the control loops to compensate for this
I did tilt my 4050 when I first installed the unit, that number is from my
memory. Notice that Magnus is right of course, the Cs will compensate the
tilt, given enough time to do so. I did not mean that the 1E-09 error will
stay permanent of course, as I said in the follow-on sentence.
Also the 4050 is not very well temperature compensated compared to modern
units, I was not impressed when I saw the FTS tempco specifications. A good
GPSDO can outperform the 4050.
I have a military OCXO sample (new product) that is supposed to be
"g-insensitive" and even that one has about E-09 frequency shift per 90
One of the only ways to get around that is to use three crystals in series,
with the three crystals oriented in the the XYZ axis, so their errors
Vibration is also a big enemy of crystal oscillators of course.
************************************** Get a sneak peek of the all-new AOL
time-nuts mailing list -- time-nuts at febo.com To unsubscribe, go to
and follow the instructions there.
More information about the time-nuts