[time-nuts] ? phase comparison or other device
cfharris at erols.com
Sat Jun 30 19:27:18 EDT 2007
Dr Bruce Griffiths wrote:
> ); SAEximRunCond expanded to false
> Errors-To: time-nuts-bounces+cfharris=erols.com+cfharris=erols.com at febo.com
> Pete wrote:
>> ); SAEximRunCond expanded to false
>> Errors-To: time-nuts-bounces+bruce.griffiths=xtra.co.nz+bruce.griffiths=xtra.co.nz at febo.com
>> This discussion is why I have allowed the ADA4899-1 to saturate in the ZCD
>> posted earlier. The transition times up & down in the second stage are 350
>> 400 ns; thus I believe the <50ns overload recovery doesn't degrade the
>> measured jitter. With 500MHz 'scope bandwidth, the 1KHz squarewave
>> output has <1% over/under shoot & appears noise free at the zero crossing.
>> The output transitions are fast enough for my HP 5370A to show < 1ns
>> jitter over 1K or 10K samples.
>> Pete Rawson
> What about the power supply noise that appears at the amplifier output
> during saturation?
> Power supply noise is almost always much larger than the opamp input noise.
> A simple resistor + diode clamp in addition to the feedback clamp will
> easily attenuate the output noise of an opamp with diode clamps across
> its feedback resistor.
The output of an opamp is always referenced to its power supply,
regardless of the configuration. If the power supply is noisy,
the output is noisy.
More information about the time-nuts