[time-nuts] Comparing Reference Accuracy
bruce.griffiths at xtra.co.nz
Thu May 22 17:26:43 EDT 2008
Stanley Reynolds wrote:
> Block diagram of using a modified version of the DO controllers to compare several disciplined oscillators. The single logging PC could be several PCs as a way to have enough serial ports.
This scheme is fatally flawed.
1) The various GPSDOs will only be statistically independent for short
2) For such short averaging times the monitoring scheme has inadequate
resolution unless the short term stability of the oscillators is very poor.
Without an independent standard of accurately known frequency such
intercomparisons can only measure the relative short term instabilities
of the various oscillators.
The errors of the various GPSDOs cannot be determined without access to
a primary standard.
Common View and All in View GPS code and GPS carrier phase measurements
may be used to compare the long term performance of the GPSDOs with a
primary standard if someone with a primary standard is logging the same
data using their primary standard and making it available.
For short term relative stability measurements a much higher resolution
scheme is required.
The SNR is likely to be sufficiently high to permit picosecond or even
subpicosecond measurement of the relative phases of the various
More information about the time-nuts