[time-nuts] HP 5372a error 160
frledda at verizon.net
Fri Sep 4 13:38:27 UTC 2009
You need to replace the internal battery (3.6) and calibrate the inputs; the
procedure is simple.
From: time-nuts-bounces at febo.com [mailto:time-nuts-bounces at febo.com]On
Behalf Of Norman J McSweyn
Sent: Friday, September 04, 2009 8:16 AM
To: Discussion of precise time and frequency measurement
Subject: [time-nuts] HP 5372a error 160
Bid on and got an HP 5372a. Supposed to be tested, working. Powers up to
error 160 out of sensitivity calibration. Did download the svc man from
Agilent. Spent a few minutes looking through the manual. Not sure what
to make of it.
It's not obvious whether a complete cal needs to be done.
Questions, comments cheerfully received.
Will be on a boat trip for our holiday (US) from 10z Saturday until
Tuesday 10z or so.
time-nuts mailing list -- time-nuts at febo.com
To unsubscribe, go to
and follow the instructions there.
More information about the time-nuts