[time-nuts] Solartron 7150plus DMM Failure
alan.melia at btinternet.com
Thu Sep 24 16:04:07 UTC 2009
Hi all, I dont know whether it is pertinent but it may pay to collect dta on
this one as there are fairures reported in many makes of equipment. the
failing part in my kit has a datecode (??) of 8922F on the paper label on
the side. By my reckoning that would be week 22 of 1989, which ties in with
the equipment production periods. I believe these fails would have been
----- Original Message -----
From: "Roy Phillips" <phill.r1 at btinternet.com>
To: "Discussion of precise time and frequency measurement"
<time-nuts at febo.com>
Sent: Thursday, September 24, 2009 10:27 AM
Subject: Re: [time-nuts] Solartron 7150plus DMM Failure
> Do you know if the Solartron 7060 (I think of the same vintage), has the
> same problem ?
> ----- Original Message -----
> From: <GandalfG8 at aol.com>
> To: <time-nuts at febo.com>
> Sent: Wednesday, September 23, 2009 10:49 PM
> Subject: Re: [time-nuts] Solartron 7150plus DMM Failure
> > In a message dated 21/09/2009 21:53:35 GMT Daylight Time,
> > lists at philpem.me.uk writes:
> > I suspect if I ever get another 7150+, I'll be swapping the line filter
> > on the instrument's arrival...
> > -----------
> > I'd go along with that.
> > Given that I'd already been advised after my first one blew that it
> > wasn't
> > unusual, I'm still wondering why I was daft enough to power up this
> > then walk away and leave it running, without some prior surgery:-(
> > regards
> > Nigel
> > GM8PZR
> > _______________________________________________
> > time-nuts mailing list -- time-nuts at febo.com
> > To unsubscribe, go to
> > https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> > and follow the instructions there.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the time-nuts