[time-nuts] X72 Rb units, any experience ?
lists at rtty.us
Mon Aug 22 21:58:48 UTC 2011
The X72 may run a DDS to do it's output.
On Aug 22, 2011, at 4:18 PM, Magnus Danielson wrote:
> On 22/08/11 22:07, Poul-Henning Kamp wrote:
>> In message<BLU125-W414D2DF78A487E3AFF8ADCE2F0 at phx.gbl>, Mark Sims writes:
>>> I have a couple of the horrid little beasties... the sine wave
>>> output is really nasty.
>> Nasty in what way ? Noise ?
> Reminds me. Friends of mine had problem with a new rubidium. They had more jitter than the phase-noise motivated. It prooved to be the 100 MHz pumping of the rubidium lamp that broke through, which was obvious when scoping the signal. It wasn't a X72, but it was a small rubidium.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts