[time-nuts] True Position Time Figure of Merit
kb8tq at n1k.org
Tue Oct 17 22:36:50 EDT 2017
First thing to check with any GPSDO is the number of sat’s in view. If it does not typically run
6 or more, the survey will take a long time.
> On Oct 17, 2017, at 6:03 PM, Stephan Flor via time-nuts <time-nuts at febo.com> wrote:
> Hi,I have been trying out my True Position GPSDO, and I have a few questions. I have been running the survey for about 32 hours, and the figure of merit goes between 2 and 4, usually 3 or 4, (range of 1-7). I have been monitoring the frequency adjusting voltage, and it varies between 2.1800 and 2.1900 v( 0-4v range). My unit has the Bliley oscillator.
> Does it have to be doing a survey to adjust the frequency? Is this behavior normal? How long should I expect to wait for a figure of merit of one?
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the time-nuts