[time-nuts] GPSDO diurnal variation
docdailey at gmail.com
Wed Aug 1 00:44:05 UTC 2012
I am not sure. I am on vacation now. Will watch it as I get. Back to normal temperature and try some other tests to sort it out. Will send you some data later.
Sent from my iPad
On Jul 31, 2012, at 5:39 PM, "Tom Van Baak" <tvb at LeapSecond.com> wrote:
> See: http://www.leapsecond.com/pages/tbolt-8d/
> As you can see, the phase stays within 5 or 10 ns during the day. The frequency stays within 5e-11 with 10 second averaging times; 5e-12 with 1000 second averaging times.
> Send me your raw data (off-list) and I'll have a look at it. How sure are you that the 2.5e-11 variation is the GPSDO and not the "precision oscillator", or your comparator?
> ----- Original Message -----
> From: "Bill Dailey" <docdailey at gmail.com>
> To: "Time Nuts" <time-nuts at febo.com>
> Sent: Tuesday, July 31, 2012 5:27 PM
> Subject: [time-nuts] GPSDO diurnal variation
>> Does anyone have plots of true frequency variation of their 10 MHz standards over days? I am trying to see if what I am seeing is typical or if I need to get a better antenna/gpsdo tuning/temp control. Right now I am on vacation so am capturing data under much warmer than normal temps (ac turned down) and am seeing 250 uHz deviation with a period of about 24 hours (2.5x10-11). I am trying to look at this in a way that is more intuitive and concrete to me than Allan deviation. I am using a precision oscillator as the comparator and am removing linear drift (stochastic) in excel. It seems surprisingly high.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the Time-nuts_lists.febo.com