[time-nuts] HP 58503A
dekagoon at gmail.com
Fri Aug 3 17:25:37 UTC 2012
I can state that my measuring of the signal (30 mV, 65,2 kHz) was caused by
the serial line of the com port of the receiver... :-)
After looking for sources of the noisy signal I disconnected my first the
GPS antenna signal from the 58512A GPS amplifier. -> Signal intact
Then I disconnect my frequency meter from the 10 MHz output -> signal intact
Now I disconnected the serial connection to my notebook -> signal gone!
So it was an interspersed signal from the serial connector ...
But at the end the question what the ESEC output may be is still mysterious...
More information about the Time-nuts_lists.febo.com