[time-nuts] 10MHz Rubidium reference source for frequency counter
brooke at pacific.net
Mon Oct 27 20:26:40 UTC 2014
The SRS PRS-10 is based on their 10 MHz SC-10 oscillator. This is what I installed in the Gibbs rack box and made use
of it's the power supplies and added down counters to get a 1 PPS output.
There are options for phase noise and aging and I suspect these same options could be used when ordering the PRS-10.
Maybe Tom has data on some of the SC-10 oscillators?
Charles Steinmetz wrote:
> Karen wrote:
>> > Really I am looking for a source with a good short range stability performance between 1 and 10 seconds.
> I concur with everything Tom wrote. If that is your goal, the best you are likely to do is with a good, free-running
> (non-disciplined) OCXO. That will have the best stability you can get (short of a hydrogen maser) from 0.1 seconds to
> ~100 seconds.
> For the convenience of having automatic drift correction, many of us use GPSDOs and turn off the disciplining before
> making measurements.
> I might consider a PRS10 as an alternative, but certainly no other "compact" Rb, disciplined or otherwise. Even the
> rack-mount HP 5065 can't beat a good free-running OCXO. Many of the "compact" Rb units (and especially the FE5680)
> have nasty output waveforms with many spurs.
> Best regards,
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the Time-nuts_lists.febo.com