[time-nuts] Generating a solid PPS from 10Mhz source
hmurray at megapathdsl.net
Fri Jan 15 04:49:17 UTC 2016
> My test methodology was to feed 10 MHz from one of my GPSDOs into the TIA
> reference and the same 10 MHz into the device input. I then had the TIA
> perform period measurements with a 10 second gate. The result was 1 second,
> ± maybe a dozen or two ps on average, but again, thatâs where the limits of
> my TIA (Keysight 53220A) lie.
Another approach is to use the delay option on a scope. Trigger on your PPS,
then look at the next one.
You probably need to do a sanity check on the scope (aka crudely calibrate)
by looking at a known-good PPS.
These are my opinions. I hate spam.
More information about the Time-nuts_lists.febo.com