[time-nuts] sine to square wave circuits - performance data?
bruce.griffiths at xtra.co.nz
Wed Oct 4 02:40:52 UTC 2017
Apparently I've already measure the PN of a comparator driving a few ACMOS buffers.
Attachment (slightly over 100kB) compares this with HX2410(Holzworth).
> On 04 October 2017 at 14:41 John Miles <john at miles.io> wrote:
> > >
> > Do you recall what the amplitude of the input signal to the Wenzel shaper
> > was?
> > Since I used a 1:2 (3dB) terminated splitter for my measurements the input
> > to the splitter is the same as the Timepod reference signal amplitude
> > >
> I was using a similar setup, so the shaper input would have been +4 to +5 dBm. My intention at the time was probably to feed it a signal comparable to the lowest levels commonly seen from typical 10 MHz sources. It may or may not do better at higher input levels, but it'll obviously get worse once the level goes below a certain point.
> -- john, KE5FX
> Miles Design LLC
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
-------------- next part --------------
A non-text attachment was scrubbed...
Size: 110594 bytes
Desc: not available
More information about the Time-nuts_lists.febo.com