[time-nuts] Weird GPSDO behavior - update
robkimberley at btinternet.com
Wed Oct 4 08:58:22 UTC 2017
Is there any way you can run it on batteries? The 8AM glitch still makes me think it is power line problem.Rob
Sent from my Samsung Galaxy smartphone.
-------- Original message --------From: Skip Withrow <skip.withrow at gmail.com> Date: 03/10/2017 19:15 (GMT+00:00) To: time-nuts <time-nuts at febo.com> Subject: [time-nuts] Weird GPSDO behavior - update
Well, I think I know a little more about my GPSDO problem, but
probably have more questions now than before. Thanks for all the
replies to the first post with thoughts and suggestions.
I first tried restarting Lady Heather and doing a cold boot on the
NTGS50AA (then entering the same disciplining values). Same behavior.
I let it run over the weekend and the same behavior happened on
Saturday and Sunday morning.
So, yesterday (Monday morning) I changed the gain to the gain of the
oscillator (.0072Hz/V), damping to 1.2, and time constant to 900s. On
the attached PRS10-2 plot you can see that it quickly settled. This
morning, it looks like all is well from the plot (about an hour before
the furnace kicks in at the right of the plot). HOWEVER, when the
plot is expanded there is still funny business going on with the DAC
control voltage at the same time of day. I just think the changed
parameters limit the disturbance. The expanded plot is the attached
At this point I'm beginning to think that the NTGS50AA is the issue,
but there are lots of questions left.
1. There are various version of the NTBW50AA/NTGS50AA GPS/operating
firmware. Mine is 10.3 and I notice that it has the LH 'ro'
designation (as does the 10.4 version). The 10.5 version does not
give the LH ro notice. Maybe it behaves better with the disciplining?
I will have to give a 10.5 a try.
2. Why does the glitch occur at 8am in the morning? Will have to try
powering the NTGS50AA up at different times and see it the glitch
3. Which disciplining parameters are affected by this glitch and which are not?
4. Have other people seen this same behavior? Does it happen on
I'll update again when I have more data.
time-nuts mailing list -- time-nuts at febo.com
To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
and follow the instructions there.
More information about the Time-nuts_lists.febo.com