[time-nuts] Fast rise time pulser
paulswedb at gmail.com
Mon Oct 16 16:57:44 EDT 2017
It did make it to the post.
Question. Rise and fall times are sub 40ps is that a typo in your email.
On Mon, Oct 16, 2017 at 10:45 AM, BIll Ezell <wje at quackers.net> wrote:
> I rushed out and ordered one, and I have to say, I'm impressed.
> The test report it came with states 34.4 ps risetime / 32.13 ps falltime
> 8.6% overshoot measured with a Tek CSA 803A.
> I hooked it up to my wonderful HP 54542 2 Gsamples/sec scope and got
> exactly what I would expect, 496 ps risetime 7.6% overshoot.
> Screenshot attached, don't know if it will actually make it into the post.
> This is exactly what I need for adjusting various of my time-related
> Bill Ezell
> The day Microsoft makes something that doesn't suck
> will be the day they make vacuum cleaners.
> Or maybe Windows 10.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/
> and follow the instructions there.
More information about the time-nuts