[time-nuts] Lucent KS-24361/Z3812A GPSDO initial setup
cbrown at woods.net
Sun Oct 26 08:11:03 UTC 2014
Ahh, I read "adjust the regulated float power supply to 28 volts" as
including the protection diode as part of the PS and 28VDC straight to
Have a half built setup like this on the bench right now.
36VDC unreg supply feeding 2 LM317 regulators with 5amp diodes on output.
Once floats a series pair of small 12v gelcells at 27.2 and the other a
single at 13.6
Need to get the enclosure and the AC filters done, they will be 24/7
powering 10811B type OCXO I want to setup as a standalone ref.
On 10/25/14, 5:54 PM, Tom Miller wrote:
> I really don't disagree with you. I did say run the supply at 28.0 and use a
> diode off the supply to the battery. That would place the float voltage at
> 27.3 or so. Best would be to follow the manufacturers float service
> recommendations. Ideally it should also be temperature compensated with
> a -2.4 mV/°C slope. Not a problem with how we use these in the lab though.
> And yes to get the longest run time, just power the main unit from the
> battery as Bob suggested.
> Now I am just waiting on some DB-9 connectors. I ran out of them and the
> local RatShak went TU.
> ----- Original Message -----
> From: "Christopher Brown" <cbrown at woods.net>
> To: <time-nuts at febo.com>
> Sent: Saturday, October 25, 2014 5:32 PM
> Subject: Re: [time-nuts] Lucent KS-24361/Z3812A GPSDO initial setup
>> Might want to dial that back a bit.
>> Since 12V/7ah batteries are mentioned I am assuming standard small
>> gelcell units.
>> Proper float voltage for a deep cycle is lower than a SLI type, and
>> gellcell even lower.
>> Generally 13.6 - 13.8 would be the gellcell range, with 13.6 being right
>> on for long life use of _small_ batteries. Equalizing charge wold be 14
>> - 14.2.
>> Floating a small gelcell at 13.6 (27.2) v.s. 14 can mean the diff
>> between 12 - 24 months v.s. 5 - 7 years service life.
>> On 10/25/14, 11:39 AM, Tom Miller wrote:
>>> These units are ideal for powering via a float charged lead acid battery.
>>> Use two 12 volt / 7 AH batteries in series and adjust the regulated float
>>> power supply to 28.0 volts. Be sure to use a diode from the supply to the
>>> battery just in case the supply can't be back fed during a power fail.
>>> ----- Original Message -----
>>> From: "Bob Camp" <kb8tq at n1k.org>
>>> To: "Bob Stewart" <bob at evoria.net>
>>> Cc: "Time Nuts" <time-nuts at febo.com>
>>> Sent: Saturday, October 25, 2014 3:01 PM
>>> Subject: Re: [time-nuts] Lucent KS-24361/Z3812A GPSDO initial setup
>>>> Spend the effort, get an antenna outside the house. Beg / borrow / steal
>>>> UPS. Even a brand new one is less than you paid for the 3812.
>>> time-nuts mailing list -- time-nuts at febo.com
>>> To unsubscribe, go to
>>> and follow the instructions there.
>> time-nuts mailing list -- time-nuts at febo.com
>> To unsubscribe, go to
>> and follow the instructions there.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to https://www.febo.com/cgi-bin/mailman/listinfo/time-nuts
> and follow the instructions there.
More information about the Time-nuts_lists.febo.com