[time-nuts] PRS10 C900 replacement
Dan Harboe Burer
dhb at post.tele.dk
Wed Jan 20 04:32:05 UTC 2016
You need one more piece of info: its value! E.g. 1uf, 0.35uF, 2.2uF, 10uF?
That is quite important :)
Just don't use too old tantalum caps - they tend to absorb moisture over
time (when stored incorrectly) - and self destruct as you have experienced..
I just experienced the same while renovating a Tek 475 'scope.
Dan (A newbie from Denmark)
Sent: Tuesday, January 19, 2016 11:21 PM
To: time-nuts at febo.com
Subject: [time-nuts] PRS10 C900 replacement
While still appearing to function normally, my PRS10 started to smell:
C900 in the lamp assembly has self destructed.
The SRS parts manual merely says "Capacitor, Tantalum, 35V, 20%, Rad",
so I'd appreciate any advice on what additional parameters I should be
looking at in selecting a replacement.
time-nuts mailing list -- time-nuts at febo.com
To unsubscribe, go to
and follow the instructions there.
More information about the Time-nuts_lists.febo.com