[time-nuts] Z3801 Newbie Question
Graham / KE9H
ke9h.graham at gmail.com
Fri Jan 29 16:15:34 UTC 2016
Your antenna setup sounds reasonable.
You might want to try dropping your elevation mask down to 5 or 10 degrees,
as an experiment.
Things that cause the sporadic jumps could be certain satellites being
added to, or removed from the solution.
Also, if there are particular angles/directions where you receive the
satellite signal as a reflection, rather than a direct signal, you get
position errors that move into and out of your solution. If the satellite
is behind one of your trees, and the received signal bouncing off the other
tree is temporarily stronger, you have a temporary path length/position
On Fri, Jan 29, 2016 at 3:54 AM, Artek Manuals <Manuals at artekmanuals.com>
> Symmetricom 58532A antenna, just above the roof, with 8 feet of RG213.
> Flat ( Florida) rural location with 360 degree view other than some trees.
> The GPS Antenna plot from Z38xx has filled in pretty much 360 degrees down
> to 20 degrees solid and in most directions to down to 15 degrees elevation.
> The exception of course is the polar abyss to the north. I am going to
> lower the elevation angle for a while just to see if there is any difference
> Looking at the link sent to me earlier
> http://www.realhamradio.com/GPS_websites_list.htm . I am seeing the same
> phenomena on most all oft the plots for the Z3801 around the world, So I
> assume that my unit is :behaving normally" for a Z3801A ... Just cant
> explain why all of us see a sporadic jumps in PPS error for an hour or two
> once or twice a day
> A man with a watch always knows what time it is
> A man with two watches is never quite sure
> . On 1/28/2016 6:23 PM, Graham / KE9H wrote:
> Please describe your Antenna setup and location.
>> How much of the sky can it see, in which directions.
>> Which directions are blocked for direct line of sight, and must depend on
>> reflections from nearby buildings or objects?
>> --- Graham
>> And here is the question is the sudden jump in the pps deviation due to
>> ionospheric and the satellite constellation change? IS there anything I can
>> do to minimize it (or is it even necessary to worry/ wonder about) This
>> behavior has going on since the beginning so it is not likely related to
>> events earlier in the week. But that set me to wondering could I minimize
>> it by say raising the elevation mask? Are there satellites in the
>> constellation that are notorious for noise etc that could be excluded from
>> the tracking list ...etc etc
>>>> A man with a watch always knows what time it is
>>>> A man with two watches is never quite sure
> This email has been checked for viruses by Avast antivirus software.
> time-nuts mailing list -- time-nuts at febo.com
> To unsubscribe, go to
> and follow the instructions there.
More information about the Time-nuts_lists.febo.com